- Vatech 3D Viewers
- Hardware Sensors & EzRay Air P
- Hardware Capture Program
- Legacy Devices
- General Q&A
Please use the following search box to quickly locate valuable FAQ support guide.You can use key words or full statement to search by. Alternatively you can simply browse by Category below
Can I use a "UPS" Battery backup to protect my X-ray device?
No - Vatech does not recommend using a "UPS" or known as Battery backup on your X-ray machine. These devices are run off battery and typically can not keep up with the power demand; typically when used will cause a long tone or beeping sound during capturing which indicates it is not able to keep up with the voltage/amperage requirements of the device.
Please view the following topic for Vatech Recommended devices.
Notice: The usage of such devices will cause damage to the equiptment and possibly void manufacture warranty.