Platform 2 (NEXGen Devices) All Listed Units are supported for upgrade to EZDent-i for 2D/3D units and Ez3D-i for all 3D Units* Pax-i Pax-i3D Pax-i3D Green Smart3D EzSensor Multi EzSesnor-Preimium EzSensor- HD *upgrade for Easydent4 sites are not included in lifetime updates. Please contact your local Vatech Rep for more information about upgrading to NEXGen […]
Vatech America does not recommend or endorse any cleaning product in particularly. It is the responsibility of the practitioner to determine what is adequate for use to clean and sanitize based on FDA as well as local codes. However based on performance and recommendation by Vatech Users ; Clorox Brand* Hydrogen peroxide wipes. (Contains no […]
Typically with in the first 5 minuets should be adequate, however industry standard is 10-15 minuets to allow for proper warmup; oils; or other parts to warm up and stabilize. Please use your best discretion.
The most common mistake is the emergency stop switch, locate the emergency stop switch located typically under the “Handle frame” this is the part of the arm that sticks out for patient to hold onto and supports the chin rest.Rotate the switch in the direction indicated by the arrows; if engaged typically it will be […]
First you will want to determine if your Legacy Device is eligible for updating to either Windows 7 or Windows 10. Click here for Legacy Windows 7 Compatibility Click here for Legacy Windows 10 Compatibility Once you have determined if your Legacy Device is eligible either of the following choices might apply A. If your […]
All Listed Units are considered “Legacy” and are non -supported for Windows 8/10. Ewoo Legacy Units Picasso Trio Picasso Duo Picasso Master3D/Master3Ds (non Picasso) Master3D/Master3Ds AnySensor (all sizes) Vatech Legacy Units: Pax500 Pax P&P PaxPrimo* (LCD TYPE) PaxFlex PaxFlex3D PaxUni , PaxUni3D (all FOV’s and versions*) PaxDuo3D (All Generation) – these will use the “LVDS” […]
Error in program callers code / CVDACQ_ETERR program Code is occurred when communication between PC (Capture program) and One Shot* Ceph sensor (OS PRO) or (OS PRO TROY) style sensor is not established, or sensor not ready state in IP communication occurs. First you will want to turn off the X-ray device for […]
Supported Windows 7 Devices: Note: All 3D Devices must be inspected for supporting “FrameGrabber/AnyGrabber” Card. IT or Certified Dealer Technician Visit is required to inspect the card to confirm it is compatible for upgrade. Vatech Legacy Units: EzSensor (Non Multi Series) Pax500 Pax P&P PaxPrimo* (LCD TYPE) PaxUni , PaxUni3D (Except H-Pannel Versions) -32bit only […]